Airflow and dynamic circumference of abdomen and thorax for adults at varied continuous positive airway pressure ventilation settings and breath rates

Continuous positive airway pressure (CPAP) ventilation is a commonly prescribed respiratory therapy providing positive end-expiratory pressure (PEEP) to assist breathing and prevent airway collapse. Setting PEEP is highly debated and it is thus primarily titrated based on symptoms of excessive or insufficient support. However, titration periods are clinician intensive and can result in barotrauma or under-oxygenation during the process. Developing model-based methods to more efficiently personalise CPAP therapy based on patient-specific response requires clinical data of lung/CPAP interactions. To this end, a trial was conducted to establish a dataset of healthy subjects lung/CPAP interaction. Pressure, flow, and tidal volume were recorded alongside secondary measures of dynamic chest and abdominal circumference, to better validate model outcomes and assess breathing modes, muscular recruitment, and effort. N = 30 subjects (15 male; 15 female) were included. Self-reported asthmatics and smokers/vapers were included, offering a preliminary assessment of any potential differences in response to CPAP from lung stiffness changes in these scenarios. Additional demographics associated with lung function (sex, age, height, and weight) were also recorded.

Diagnostically, chronic respiratory diseases are classified by the composition and location of obstructive and restrictive abnormalities 8,22 . Obstructive abnormalities act to occlude airways through constriction, inflammation, or collapse and cause increased airway resistance 8,23,24 . Restrictive abnormalities can be intrinsic and stiffen lung tissue by scaring/fibrosis or extrinsic and stiffen the thorax in cases of severe burns and obesity 3,8,[25][26][27] . Clinically, the potential presence of abnormalities is primarily indicated by observed changes to breath patterns, palpation, percussion, auscultation and oximetry 8,[28][29][30] .
Continuous positive airway pressure (CPAP) ventilation is a common respiratory therapy in both clinical and home settings 31,32 . CPAP provides a set positive end-expiratory pressure (PEEP), which reduces the patient work/effort required to generate inhalation by increasing the effective pressure gradient from expansion 33 . PEEP also maintains airway integrity, which can be impacted in cases such as obstructive sleep apnoea (OSA), preventing airway collapse 31,33,34 . The outlined trial was conducted to include a range of tested PEEP settings to understand the interaction between lung mechanics and CPAP using healthy subjects.
Pressure and flow data was recorded at the CPAP mask interface (mouth/nose orifice), from which tidal volumes were computed (Fig. 1). To elucidate thoracic and abdominal breathing modes circumferential monitoring was included in this trial (Fig. 1). These measures were taken at both chest and abdominal levels to better differentiate breathing modes ( Fig. 1) 35 . This data also provides a second measurement set to validate and inform pressure and flow-based modelling.
The purpose of this dataset is to inform the development of model-based methods of respiratory assessment and treatment setting protocols. To personalise and optimise remote respiratory care, model-based methods must be developed to assess respiratory function including obstructive and restrictive abnormalities, signs of distress, and response to non-invasive mechanical ventilation (i.e. CPAP). Model-based methods have been used to identify patient effort, lung elastance, and airway resistance, with and without MV support [36][37][38][39][40][41][42][43][44][45] . This trial specifically examines the impact of CPAP on healthy lungs and respiration to capture the fundamental dynamics. Dynamic circumference measurements provide further information about muscular contributions to effort and different breathing modes.

Methods
Ethical consent for the trial was granted by the Human Research Ethics Committee (HREC) at the University of Canterbury (HEC 2020/14/LR). Participants consented to the open publication of their de-identified data.
Experimental design. The trial was designed for a cohort of N = 30 subjects, with an even split of males (15) and females (15), and including self-reported asthmatics and smokers. People with heart conditions or other serious medical conditions were excluded from participation. Table 1 outlines the trial procedure proposed to capture data from three breath rates (panting, normal, deep breathing) over three PEEP levels (ZEEP (zero end-expiratory pressure), 4 and 8 cmH 2 O).
Breathing was un-cued (without a metronome) to capture as many breath modes as possible within the cohort in response to verbal instructions to pant as fast as possible, breathe normally, and breathe as deeply as possible. The trial was designed to include 4cmH 2 O PEEP, as it is a commonly prescribed initial or minimum PEEP level for patients during CPAP therapy [46][47][48] . A control level at ZEEP (PEEP = 0 cmH 2 O) was chosen to capture natural unassisted mechanics. A higher support level was subsequently set to 8cmH 2 O to provide a uniform level of increased support. All breathing rates were assessed at each PEEP level, with subject-determined breaks between each to mitigate fatigue. Fig. 1 Data measurement illustration. The subject's were semi-prone with abdominal circumference taken at the waist, chest circumference taken at the armpit level, and Pressure and Flow measured at the mask interface. An example of this collected data is illustrated for a single breath.
Overall, the trial was designed to capture a broad range of mechanics to inform the design of more extensive trials aimed at improving the prescription of CPAP therapy, which currently has varied efficacy and adherence across different demographics 49,50 .
At the start of the trial, prior to obtaining informed consent, the researcher described the procedure to the subject, and provided them with a written summary. After consent was obtained, demographic data was collected using a questionnaire completed at the start of the trial, measurement equipment was made available for height and weight. Subsequently, a CPAP mask was fitted to the patient and the respiratory circuitry was connected.
Data acquisition. Data were recorded using a custom Venturi-based bidirectional differential pressure sensor array (Fig. 2) and rotary encoder-based dynamic circumference tapes (Fig. 3). Details of the design and validation of these open-design measurement devices, including links to open-access design files, is available at Mendeley Data 51 . Data was collected using Matlab (Matlab 2021b, The Mathworks Inc, Natick, MA, USA), sampling at 100 Hz. The differential pressure over the venturi constriction was recorded in both directions and between the throat and atmosphere (gauge) (Fig. 2), alongside thoracic and abdominal circumferences (Fig. 3).
Data processing. Data was processed in MATLAB, with both raw and processed data available in the repository. Differential pressures were converted to cmH 2 O units based on datasheet transfer function information, defined: The differential pressure sensors (P1J-10-AX16PA) had a pressure range of 0-10"H 2 O (0-25.4 cmH2O), corresponding to a digital count of 1638 at 0% Pressure (10% of 2^14 counts) and 14745 at 100% Pressure (90% of 2^14 counts).
The venturi differential pressures were combined, to read the differential pressure over the intake with respect to the direction of flow, and used to compute the flow: Where, A 1 and A 2 are cross-sectional areas at the intake and constriction, respectively (d 1 = 15 mm and d 2 = 10 mm). Given a discharge coefficient, c d = 0.97, and the density of air, ρ = 1.225 kg/m 3 . Inspiratory start points were identified, and tidal volumes, V [L] were computed as the integral of flow, Q [L/s], over time, t [s], which is zeroed at each inspiratory start index. Circumferences were measured using rotary encoders measuring revolutions of a tape barrel as the tape was unspooled and spooled during expansion and contraction, respectively, of the chest or abdomen 51 . Tapes captured the initial circumference during the unspooling of the tape to fit the subject 51 . Circumferences (C) were converted from rotary counts based on the initial unspooled length (C 0 = 108 mm) and encoder counts per revolution (4096) 51 , yielding: Given the tape barrel radius (r), calculated from the number of complete barrel revolutions, the initial radius of the full spool (r 0 = 22 mm), and tape thickness (t t = 0.15 mm):   www.nature.com/scientificdata www.nature.com/scientificdata/ start indices, by identification of a change in flow direction from expiratory to inspiratory, and considered to end at the point prior to the next inspiratory start index. An example of a processed dataset in Fig. 4, against time [s] with inspiratory start indexes indicated with vertical dashed lines 52 .

LICENSE.txt
Data licence (Creative commons attribution 4.0 international public license).

README.txt
Data descriptor file containing an explanation of file organisation and contents.

subject-info.csv
Spreadsheet of self-reported medical information for the 30 subjects, classified by subject number. www.nature.com/scientificdata www.nature.com/scientificdata/ Raw datasets are also included as ADC direct output files and as files with processed units but differential pressures unresolved into the bidirectional flow (Table 2). Files are saved in 'raw-data.zip' under folders by subject number ('01' through to '30'). Files are saved in these folders by PEEP setting and breath rate type (Subject Number is also referenced).
Demographic data is included in a spreadsheet 'subject-info.csv' ( Table 2). Spreadsheet columns A to M, respectively, contain subject number, sex (M/F), height [cm], weight [kg], age [years], history of smoking and/ or vaping (Yes/No), smoking/vaping frequency (units included in column values), duration of smoking (units included in column value), history of asthma (Yes/No), asthma medication name (if applicable), frequency of use of asthma medication (if applicable, units included in column value), any trials the subject perceived to make it harder to breathe (PEEP setting [cmH2O]), and any trials the subject perceived to make it easier to breathe (PEEP setting [cmH2O]).

Technical Validation
The hardware system validation is outlined 51 . Flow was calibrated against a TSI 4000 Series externally calibrated flow sensor 51 . Tape extension accuracy was assessed by extensions to a series of known values between 200 and 1500 mm 51 .

Usage Notes
A prevalent issue in all PAP and especially CPAP therapy is mask leakage [53][54][55] . Efforts were made to reduce leakage in this trial by ensuring the mask fit very tightly at the start of the test and asking the patient to signal to the researchers if they feel any leaks around this seal. However, during preliminary analysis, volume deficits indicative of mask leakage were noted in the data from some trials. Mask leakage is a consideration when using data to identify expiratory patient-specific lung mechanics by model-based methods. Uncaptured, leakage flow may impact model parameters identified during expiratory periods and thus the accuracy of model-fit to the breath. Expiratory mask leakage affect only measured expired flow (hence volume), and so peak tidal volumes and inspiratory volumes and flow profiles were unaffected. Dynamic circumference data was captured independently, and not impacted by mask leakage.
It should also be noted gauge pressure measurements hit sensor maximum in some trials during high flow rates, predominantly seen in panting. This issue did not impact other measurements in the trial. However, in the application of models to this dataset, regions of saturated peak pressures should be considered and not be used to identify pulmonary mechanics parameters. In future trials, the gauge pressure sensor port will be relocated from the constriction and higher-range sensors implemented.

Code availability
Data was collected using MATLAB code and custom hardware. Design files and code are available openaccess, with manufacture and usage instructions published in a HardwareX article 51 . The collected data, subject demographic information, and figure generation code are published in a Physionet repository 52 .